1. Find the calibration software Open the EzCad2 file directory and double-click “Corfile2” to enter the 9-point calibration software. 2. Set the calibration parameters Click “F1 设置激光器参数” to set the laser type and marking parameters (normally, the marking parameters are default). In general, the fiber laser type is IPG, SPI, and is checked separately. Click… Continue reading EzCad2 Correction File: 9-Point Calibration Manual
After replacing the galvanometer, did the center point position change?
After the replacement, it was found that the positions of the text and graphics in the previously saved file is not the same as before. After changing the galvanometer, do the galvanometer calibration again.
What’s wrong with the prompt “Can not open correct file” after opening the software EzCad?
It is because there is no correction file and the “Use the correction file” in the F3 parameter is checked. If there is a correction file, please select the correct path; if there is no one, you can leave “Use the correction file” unchecked. Related: How to set Ezcad2 correction file?